Additel ADT161 & ADT161EX Intelligent Digital Pressure Modules

OVERVIEW

ADDITEL NOTICE: ADT160A pressure modules will be REPLACED by the models ADT161 &
ADT161Ex (intrinsically safe) models

Although the specifications are identical between the ADT160 and new ADT161 series of
pressure modules, this change allows us to provide a complete intrinsically safe version
(ADT161Ex) of pressure modules that can be coupled with our new ADT226Ex & ADT227Ex
Multifunction Process Calibrators. Legacy ADT160A pressure modules can be used with the new ADT226 & ADT227
calibrators, but are not compatible with the new intrinsically safe ADT226EX &
ADT227Ex calibrators.

With advanced microprocessor technology and state-of-the-art silicon pressure sensors, Additel’s 161 and 161Ex series
Digital Pressure Modules provide an accurate, reliable, and economic solution for wide range of pressure applications.
Our intrinsically safe (Ex) models are up to the task of providing the best possible results, even in hazardous
environments. In order to reach the best performance, every silicon pressure sensor in the module has been specially
aged, tested and screened before assembly. Designed as external pressure modules for Additel’s 760 automatic
handheld pressure calibrator, the ADT761 automated pressure calibrator, ADT226 series handheld process calibrator
and Additel’s flagship 780 pressure controller, the Additel 161 is unmatched in performance and reliability. If intrinsic
safety is a critical requirement for your workload, we have you covered when you combine our model ADT226Ex
process calibrator with any of our ADT161Ex pressure modules.

GENERAL FEATURES

• Gauge pressure measurement accuracy of 0.02% FS
Absolute Pressure measurement accuracy of 0.1% FS

• Pressure ranges to 60,000 psi (4,200 bar)

• Advanced temperature compensation

• NIST traceable calibration and data included

Gauge Pressure [1]

P/N

Pressure Range

Media [2]

Accuracy(%FS)

Burst Pressure

(psi)

(bar)

V15

-15

-1

G

0.02

3×

GP2

2

0.16

G

0.05

3×

GP5

5

0.35

G

0.05

3×

GP10

10

0.7

G

0.02

3×

GP15

15

1

G

0.02

3×

GP30

30

2

G

0.02

3×

GP50

50

3.5

G , L

0.02

3×

GP100

100

7

G , L

0.02

3×

GP150

150

10

G , L

0.02

3×

GP300

300

20

G , L

0.02

3×

GP500

500

35

G , L

0.02

3×

GP600

600

40

G , L

0.02

3×

GP1K

1000

70

G , L

0.02

3×

GP1.5K

1500

100

G , L

0.02

3×

GP2K

2000

140

G , L

0.02

3×

GP3K

3000

200

G , L

0.02

3×

GP5K

5000

350

G , L

0.02

3×

GP10K

10000

700

G , L

0.02

2×

GP15K

15000

1000

G , L

0.05

2x

GP20K

20000

1400

G , L

0.05

1.5x

GP25K

25000

1600

G , L

0.05

1.5x

GP30K

30000

2000

G , L

0.05

1.5x

GP36K

36000

2500

G , L

0.05

1.5x

GP40K

40000

2800

G , L

0.05

1.35x

GP50K

50000

3500

G , L

0.1

1.2x

GP60K

60000

4200

G , L

0.1

1.1x

[1]. Sealed gauge pressure for above 1000 psi

[2]. G=Gas, L=Liquid

Differential Pressure

P/N

Pressure Range[1]

Media

Accuracy (%FS)

Burst Pressure

Static Pressure Range

(inH20)

(mbar)

DP1

±1

±2.5

G

0.05[2]

100×

±10 psi

DP2

±2

±5.0

G

0.05[2]

100×

±10 psi

DP5

±5

±10

G

0.05[2]

50×

±10 psi

DP10

±10

±25

G

0.05[2]

20×

±10 psi

DP20

±20

±50

G

0.05

20×

±10 psi

DP30

±30

±75

G

0.05

20×

±10 psi

DP50

±50

±160

G

0.05

3×

±10 psi

DP100

±100

±250

G

0.02

3×

±15 psi

DP150

±150

±350

G

0.02

3×

50 psi

DP300

±300

±700

G

0.02

3×

50 psi

[1] FS specification applies to the span of the range. Accuracy includes 1 year stability.

[2] 0.05%FS accuracy (incl 6 months stability). One year accuracy is 0.05%FS calibration accuracy combined with 0.05%FS one year stability.

Compound Pressure

P/N

Pressure Range

Media

Accuracy

Pressure Rating

psig

bar.g

Burst

Over Pressure

CP2

±2

±0.16

G

0.05% FS

3x

1.2x

CP5

±5

±0.35

G

0.02% FS

3x

1.2x

CP10

±10

±0.7

G

0.02% FS

3x

1.2x

CP15

±15

±1.0

G

0.02% FS

3x

1.2x

CP30

-15 to 30

-1 to 2.0

G

0.02% FS

3x

1.2x

CP50

-15 to 50

-1 to 3.5

G

0.02% FS

3x

1.2x

CP100

-15 to 100

-1 to 7.0

G,L

0.02% FS

3x

1.2x

CP300

-15 to 300

-1 to 20

G,L

0.02% FS

3x

1.2x

CP500

-15 to 500

-1 to 35

G,L

0.02% FS

3x

1.2x

CP600

-15 to 600

-1 to 40

G,L

0.02% FS

3x

1.2x

CP1K

-15 to 1,000

-1 to 70

G,L

0.02% FS

3x

1.2x

CP2K

-15 to 2,000

-1 to 140

G,L

0.02% FS

3x

1.2x

CP3K

-15 to 3,000

-1 to 200

G,L

0.02% FS

3x

1.2x

CP5K

-15 to 5,000

-1 to 350

G,L

0.02% FS

3x

1.2x

CP10K

-15 to 10,000

-1 to 700

G,L

0.02% FS

2x

1.2x

Absolute Pressure

P/N

Pressure Range

Media

Accuracy(%FS)

Burst Pressure

(psi)

(bar)

AP5

5

0.35

G

0.1

3×

AP10

10

0.7

G

0.1

3×

AP15

15

1

G

0.1

3×

AP30

30

2

G

0.1

3×

AP50

50

3.5

G

0.1

3×

AP100

100

7

G , L

0.05 (0.1)

3×

AP300

300

20

G , L

0.05 (0.1)

3×

AP500

500

35

G , L

0.05 (0.1)

3×

AP1K

1000

70

G , L

0.05 (0.1)

3×

AP3K

3000

200

G , L

0.05 (0.1)

3×

Barometric Pressure

P/N

Pressure Range

Media

Accuracy

Burst Pressure

Low

High

BP

60 kPa

110 kPa

G

40 Pa

3×

Precision Sensors[1] [3] - Absolute Pressure

P/N

Pressure Range

Media

Accuracy[2]

Pressure Rating

psi.a

bar.a

Burst

Over Pressure

AP15R

15

1

G

0.01% FS

2x

1.2x

AP30R

30

2

G

0.01% rdg or 0.003% FS whichever is greater

2x

1.2x

AP50R

50

3.5

G

0.01% rdg or 0.003% FS whichever is greater

2x

1.2x

AP100R

115

8

G,L

0.01% rdg or 0.003% FS whichever is greater

2x

1.2x

AP300R

315

21

G,L

0.01% rdg or 0.003% FS whichever is greater

2x

1.2x

AP500R

515

36

G,L

0.01% rdg or 0.003% FS whichever is greater

2x

1.2x

AP1KR

1015

71

G,L

0.01% rdg or 0.003% FS whichever is greater

2x

1.2x

AP2KR

2015

141

G,L

0.01% rdg or 0.003% FS whichever is greater

3x

1.1x

AP3KR

3015

201

G,L

0.01% rdg or 0.003% FS whichever is greater

3x

1.1x

AP6KQ

6000

400

G,L

0.01% FS

3x

1.2x

AP10KQ

10000

700

G,L

0.01% FS

2x

1.2x

AP15KQ

15000

1000

G,L

0.01% FS

2x

1.2x

AP20KQ

20000

1400

G,L

0.01% FS

2x

1.2x

AP30KQ

30000

2000

G,L

0.02% FS

1.5x

1.1x

[1] Contact Additel for other range options.

[2] Accuracy includes calibration uncertainty, linearity and long-term stability

[3] Precision Sensors (APXR and APXQ) cannot be configured as an Ex model and cannot be read by Ex device.

Additel 226Ex with ADT161Ex Pressure Module

Standard Accuracy

Precision Accuracy

CPXXX

DPXXX

GPXXX

APXXQ

APXXR

Operating temperature

-10°C to 50°C (14°F to 122°F)

15°C to 25°C (59°F to 77°F)

10°C to 30°C (50°F to 86°F)

Storage temperature

-20°C to 70°C (-4°F to 158°F)

-20°C to 70°C (-4°F to 158°F)

Relative humidity

95% RH

90% RH

Pressure connections (for external use only)

1/4NPT, 1/2NPT, 1/4BSP, 1/2BPS, M20x15

1/4NPT, 1/2NPT, 1/4BSP, 1/2BSP, M20x15

Enclosure (for external use only)

SS enclosure

SS enclosure

Intrinisic Safety (ADT161Ex models only)

ATEX certified intrinsically safe II 1G EX ia IIC T4 Ga TUR 16.0023X

• Sourcing, simulating and measuring pressure, temperature and electrical signals
• Built-in Barometer
• Intrinsically safe models available (Ex)
• Large Smartphone like touchscreen user experience
• USB Type-C and Bluetooth communications

• Sourcing, simulating and measuring pressure, temperature and electrical signals
• Built-in Full Hart Communicator (ADT227-HART)
• Built-in Barometer
• Intrinsically safe models available (Ex)
• Large Smartphone Like Touchscreen User Experience